Dental composite curing light
This LED composite curing light features two working modes: High and Norm on time setting.
The dental composite curing light unit time setting:
- High: 1S, 3S,
- Norm: 5S. 10S. 15S. 20S. 25s. 30s
The solidification effect is not affected by the consumption of remaining power.
A full charge can be used for more than 400 times continuously under 10s working time mode.
Principle and Application
- adopts the principle of ray radiation to solidify the light-sensitive resin by shooting at it in a short time.
- This product is used for dentistry. It has the function of accelerating the material of dental restoration curing.
- LED curing light (dentistry) is mainly composed by LED, light hood, charging cradle, battery, adapter, main unit
Installation and Remounting
1. Aim the mounting holes which on the upper of machine, and make the long side of the top of the main unit inserted to it(must screw the top of the main unit to the end, don’t inclined inserted)
2. When the battery needs to be charged, connect the plug of the adapter into the AC100V-240V power supply. Then connect the output plug of the adapter to the DC 5V input plug of the pedestal, then put the main unit of composite curing light into the pedestal. Please pull out the adapter after charging.
Composite curing light operation
1. Strong Model: screen shows P1, press the button to select of time which could be 1s, 3s. Its output light intensity is about1200-1400mw/cm2.
2. Pulse Mode: screen shows P2, press the time button to select of time which could be 5s,10s, 15s, 20s,25s,30s. Its output light intensity is about800-1000mw/cm2.
3. Standard Mode: screen shows P3, press the time button to select of time which could be 5s, 10s, 15s, 20s. 25s.30s Its output light intensity is about800-1000mw/cm2.
4. Press the time button to set the solidifying time on the dental composite curing light unit.
- High: 1S, 3S
- Norm: 5S, 10S, 15S, 20S 25S 30S the solidifying time
5. When operating, aim the top of the main unit at the correct position, press the power button, a”Di’’ sound will appear, the LED shine the blue light and start to work under the selected mode Screen began to display the countdown time. When the countdown back to 0 the work finished. Then the screen will return to setting time.
6. While in operation, press the power switch button can stop work any time.
7. At the end of a working cycle, the next working cycle can be started immediately by short pressing the button. If the main unit gets hot, obviously, please turn off the device until the main unit becomes cool. Please don’make it continuously illuminates more than 10 times.
8. Low power detect circuit is fixed inside the main unit, when low power is detected, the battery symbol display LED flashing, please do not disassemble the Lithium battery, it will lead to the circuit short or the electrolyte leakage
9. Do not squeeze, shake and short the battery, do not store the battery with metal material.
10. The instrument has electromagnetic interference. Do not use around the electronic operation, at the same time have a strong electromagnetic interference environment should be careful to use the instrument.
11. It is forbidden to use when charging operation.
12. This product should be used by trained, qualified dentists And this product is suitable for dental patients. Must be use in hospital or professional medical site.
13. To avoid electromagnetic interference, the device should be installed at the medical site which meet the requirement of EMC WARNING: The adapter should be connected to the socket which is easy for operator to touch.
- WARNING: over-heat scorching: the device cannot be used for 20s continuously.
- WARNING: High temperature burn, The dental composite curing light unit can not direct shining to the skin tissue like lips and mucosa.
1. Rechargeable Lithium battery:
- Battery model: JW-Y-1.5, Battery capacity: 1500mAh
- Battery has over-voltage, over current and short circuit protection
- Adapter input: AC100-240V 50HZ/60HZ
- Output: DC5V 1A
- Built-in fuse: T1A250V
3. Light source:
- 1 5W high power blue light LED
- Wave length: 420 nm-480 nm
- Class: class I
Check method: When operate the dental composite curing light unit properly, LED luminous means LED is in good condition.
The wavelength of our LED curing light machine can match with dental resin material which are commonly used on clinical, such as 3M and Dentsply.
420 nm to 480 nm(blue light) wavelength range of radiation not less than 250 MW/cm2.
The heart disease patients, pregnant women and children should be cautious to use this equipment
1. This dental composite curing light unit does not include the self-maintainable spare parts. The maintenance of
this equipment should be taken by the appointed professional or special repair shop.
2. The accessory of the product should be cleaned by clean water or sterilized liquid. Do not soak.
3. Please clean the resin remained on the top of the main curing light dental unit after using to avoid infecting the life-span or solidified effect.